Air Quality - Gas contents - 2
This webpage provides the following information:
Measurement/ Detection of Gasses CO, NH3 and NO2 by sensor MiCS6814
As start a basic configuration which is rather broadband, with a combined gas sensor which in (limited) quality and (limited) sensitivity is compatible with the dust-sensor GP2Y10
Sensor type MiCS6814 is best available, already implemented on a break-out PCB
With multiple sensors it measures & reports gas level divided in 3 signals for CO, NH3 and NO2, as a kind of spectrum-survey aimed at
"detection of pollution from automobile exhausts and for agricultural/industrial odors" [quote from datasheet].
Relative to the single value measured by the companion type MiCS2714,
this latter capability of type MiCS6814 seems useful for a role as 'general alerter',
and for some 'practical learning' about the contents of the environment.
The 3 analogue signals from the MiCS6814 are read by the channels 0 till 2 of an ADC type ADS1115.
Till the availablity of a simple, local calibration & scaling, these ADC-signals initially will be scaled on the 10 bits of the ADC => 0 till 1024 units,
directly compatible with the setting for readout of ADS1115: fullscale = 1024mV => 1unit = 1mV
When calibration has been found, either the controlling ESP8266, or the next stage in Domoticz for each type of gas will translate/scale the measured amount of units into microgram/m3.
Adjacent to the gassensor an extra (simple) thermosensor LM35 measures the temperature of the air flowing over the gassensor:
the sensor LM35 is physically fitted close and hovering over the gassensor for a local measurement of temperature as close as possible.
It's analogue output is a linear voltage of 10mV/degree, read by channel 3 of the same ADS1115:
also this 3rd channel of the ADC applies the setting of fullscale = 1024mV => 1unit = 1mV => resolution = 0.1 degree
anticipating the measurement of positive temperatures.
The cluster of Gassensor+Thermosensor is free-hanging in the housing in front of the dustsensor:
no protection against temperature, moisture and dust from ambient air, as would be required for more stable measurements.
ToDo-List for the 'basic' configuration
To be investigated & to be developed:
- Calibration concept =
data logging => offsets + transfer function
to become compatible with RIVM's AQI-classsification
- Environmental control =
stable flow circuit with control for heating/drying and for filtering of the air/gas for gassensor & dustsensor
to compensate effects of temperature & humidity, respectively dust (related to gas measurements):
First feasible approach seems:
- a tube/canal in front of the entry of the GP2Y10-sensor,
- enclosing the cluster of gassensor MiCS6814 + thermosensor LM35,
- with heater at entry of tube/canal,
- with heater power [PWM?]-controlled from the LM35-info
- NO dustfilter, being too complicated to fit & maintain
[Such filter should be positioned in the air flow 'before' the gas-sensor:
however conflicting with the present position of the gas-sensor cluster in front of the dust-sensor,
would block good operation of the dust-sensor,
which is now located 'behind' the gassensor]
For the 'Extended Configuration' the concept must be developed:
still in infancy.
Back to main Air Quality webpage
Back to page for Meteo
The websection for Experiments starts here
Sitemap/ Jumplist voor deze website, incl. links to english versions of pages
Copyright © 2013-2020 T4S
The copyright for information available from or accessed via external links (also without declaration of source), remains with the original owner, hence the use of such info may be subject to limitations.